About McStas Download Documentation 
[ Identification  Description  Input parameters  Output parameters  Links ] The
Materials Probe for Internal Strain Investigations

Name  Unit  Description  Default 

source_lam_min  Angs  Minimum wavelenth of source  0.5 
source_lam_max  Angs  Maximum wavelenth of source  2.0 
hi_res    Selects hiresolution(1) or hiintensity(0) reactor beam through primary shutter  0 
mono_Si_type    Monochromator Silicon type options: 422 400 311 511 111 331  311 
mono_mosh  arc min  Monochromator horizontal mosaicity  30 
mono_mosv  arc min  Monochromator vertical mosaicity  30 
mono_dx  m  Monochromator delta x  positive to left of reactor beam  0 
mono_dy  m  Monochromator delta y  positive upward of reactor beam  0 
mono_dz  m  Monochromator delta z  positive along reactor beam  0 
mono_takeoff  deg  Monochromator takeoff angle  positive anticlockwise from reactor beam  83.5 
mono_dtilt  deg  Monochromator tilt angle  not implemented yet  0 
mono_Rh  m  Monochromator horizontal focus radius  3.572 
port_takeoff  deg  Port takeoff angle  positive anti clockwise from reactor beam  83.5 
chamber_window_rad  m  Chamber window radius. If this is 0, there is no window  0.05534 
inc_slit_rot  deg  Incident slit delta rotation  not implemented yet  0 
inc_slit_dx  m  Incident slit delta x position  positive to left of incident beam  0 
inc_slit_to_cor  m  Incident slit to sample stage center of rotation  0.005 
inc_slit_width  m  Incident slit width 0.00013m to 0.005m  0.005 
inc_slit_height  m  Incident slit height 0m to 0.02m  0.02 
inc_slit_sep    Incident slit separation between width and height. <0:use emperical calc, >=0:distance in m  1 
mono_to_cor  m  Distance between monochromator and center of rotation  2.5 
sample_dx  m  Sample delta x  positive to left of incident beam if sample_dom=0  0 
sample_dy  m  Sample delta y  positive upword of incident beam  0 
sample_dz  m  Sample delta x  positive along of incident beam if sample_dom=0  0 
sample_dom  deg  Sample delta omega  positive anticlockwise from incident beam  0 
det_takeoff  deg  Detector takeoff angle  positive anticlockwise from incident beam  90 
cor_to_det  m  Distance between sample centre of rotation and detector  1.148 
diff_slit_dx  m  Diffracted slit delta x position  positive to left of diffracted beam  0 
diff_slit_to_cor  deg  Distance between centre of rotation and diffracted slit  0.005 
diff_slit_width  m  Diffracted slit width  0.005 
diff_slit_height  m  Diffracted slit height  0.02 
full_instrument    When 1, simulates the complete instrument. When 0, only simulate from the outlet collimator  1 
SAFARI_MPISI.instr
.
[ Identification  Description  Input parameters  Output parameters  Links ]
Generated automatically by McDoc, Peter Willendrup <peter.willendrup@risoe.dk> / Fri Jan 18 10:06:10 2019